Most Popular Articles
Testing Antennas with a Network Analyzer
No doubt one of the most revolutionary tools for the RF side of broadcasting was the advent of the network analyzer. Prior to its invention, measuring the response of an antenna system was more or less limited to using a bridge or vector voltmeter, a swept spectrum analyzer with return loss bridge, or the reflected power meter on the transmitter. Each of these methods had limitations that were ultimately eliminated with the network analyzer.
Using a bridge or vector voltmeter, while reasonably accurate at measured frequencies, was very time consuming. Graphical interpolation between the measurement frequencies often induced substantial error in the measured data, although this was less of an issue for FM antennas and shorter line runs. Spectrum analyzer methodologies are quicker and provide greater resolution and less error, but are unable to present the entire story. You may be able to determine that a problem exists; however, where that problem is located typically remains a mystery. Finally, using the reflected power meter to tune an antenna is an absolute no-no. While the transmitter meter is useful for noting changes, this methodology is wholly unsuitable for any meaningful adjustment, and can mistune the antenna substantially enough to result in a combustible failure.
Network analyzers come in two flavors: scalar and vector. The scalar network analyzer provides essentially the same information as a swept spectrum analyzer with a return loss bridge, and is thus not as robust a tool as its big brother the vector network analyzer or VNA. Vector analyzers measure both magnitude and phase of the response, from which all of the important other data formats can be mathematically calculated.
The basis of the calculations in the VNA is S-parameters. S-parameters work by characterizing a network, in this case an antenna or RF system, through the use of matched loads instead of open and/or short circuit conditions. It is much simpler to characterize a resistive load across a wide range of frequencies than it is to accurately represent a true open or short circuit condition. For broadcasting applications we typically are interested in the S11 and S12 cases. The S12 case arises when examining the insertion characteristics of filters or combiners, and is the response looking from port 1 through port 2 of the system. The S11 case is the response when the reflected signal at the input is compared to the incident signal, which is the measurement mode used when viewing what is seen by the transmitter.
The stock VNA will usually only provide information in the frequency domain. A plot in the frequency domain will have frequency on the horizontal or X axis and the magnitude of a particular quantity on the Y or vertical axis. So just off the shelf we can easily view formats such as VSWR, return loss, the linear magnitude of the reflection coefficient, and phase. When the magnitude and phase are combined, we are able to look at polar return loss and the Smith chart. These latter two formats will qualitatively indicate which end of the system a problem is occurring. Indeed with some experience this information can be gleaned from the Cartesian VSWR and return loss plots. The problem of course is where exactly in the system is the problem occurring?
The time domain option available on most modern vector analyzers solves this issue. In the time domain mode, the horizontal axis of Cartesian plots becomes time, while the vertical axis remains in terms of magnitude. Since it takes a finite amount of time for a signal to propagate through a transmission line, a particular "time" of interest is directly relatable to distance down the line. The time domain information is mathematically generated through the application of the inverse Fourier transform to the magnitude and phase acquired at each individual swept frequency. The result is that through the proper selection of measurement frequencies, an issue in a system can be located with very good accuracy.
To adequately depict the condition of an antenna system both the frequency and time domain must be considered. Frequency domain data provides a view of what the transmitter is actually seeing. The time domain data allows anomalies in the transmission line and the magnitude of the far end reflection, which was very important in analog television, to be quantified.
- continued on page 2
Acceptable Use Policy blog comments powered by Disqus
[an error occurred while processing this directive]
Today in Radio History
The history of radio broadcasting extends beyond the work of a few famous inventors.
Read each issue online in our Digital Edition Format in your Web browser.
EAS Information More on EAS
The feed provides feeds for all US states and territories.
Need a calendar for your computer desktop? Use one of ours.
Browse Back Issues[an error occurred while processing this directive]
Also in the December Issue
- Local Radio Spotlight: Koser Radio Group
- Trends in Technology: Streaming Audio Update
- Contest Rules Rewrite and EAS Issues
- Embedded Computing, With a Side of Pi
- Field Report: TASCAM US-366